No. If you are comparing the latest high performance white and green phosphor image intensifiers then they will have the same performance measurements. The color comes from the application of white or green phosphor after the conversion of photons to electrons and the multiplication of those electrons within the image intensifier. The color of the phosphor may offer a variation in contrast sensitivity which may be perceived differently among each individual creating their own pros and cons....
72 hours (3 business days) pending the system does not require any additional major repair. Typical turnaround has been 24 to 48 hours.
The NVG and the battery pack are the only two required items to be inspected every 180 days but we recommend sending in the helmet mount as well so we can verify the entire system functions properly as a faulty helmet mount can render the entire system inoperative.
Historical record review, inventory components, clean eyepiece and objective lens, clean overall housing, verify range of adjustment limitations, verify proper function of low battery light and switch mechanism provides power to each side of battery pack , helmet mount continuity check, image intensifier deficiency and performance checks, adjust collimation, zero diopters and purge with nitrogen if necessary.
No, you do not need an RMA for 180 day inspections but you need to either go to our website and fill out an 1141 Shipping Form and electronically submit or print and send in with the NVGs.